Surface Error of Convex Optical Surfaces
Verifying the surface quality of convex lenses, mirrors, domes, and telescope secondary optics is typically accomplished with a Fizeau Laser Interferometer. A short coherence laser interferometer can also be used, and it is the preferred option for measuring thin domes.
Measure Convex Surface Error with a Fizeau Interferometer
Standard setup for measuring the surface quality of a convex optical surface.
|AccuFiz Interferometer||More Information|
|Bayonet or 3-Jaw Mount||Specifications|
A vertical configuration can speed measurements, particularly for repeated measurements of the same type of component. The lightweight AccuFiz interferometer and lookdown stand make vertical measurement easy.
Measuring High Reflectors
For highly reflective optics the test beam returned to the system may be so bright that it washes out the interference fringes. A beam attenuator such as a pellicle can be inserted into the test path to improve beam contrast. A reference optic with attenuation coating can also be used.
A pellicle improves fringe contrast with a highly reflective optic.
Measuring in Vibration
With a traditional, phase-shifting interferometer, measurement data is acquired sequentially—a long enough process that vibration and air turbulence will degrade or even prevent measurement. The interferometer and all mounts must be situated on an air table to isolate the system from vibration. Air isolation can be expensive—prohibitively expensive for very large optics or long measurement paths.
4D AccuFiz interferometers feature optional Dynamic Interferometry®, which enables all measurement data to be acquired simultaneously. Vibration is no longer an issue, and an isolation table is not needed.
Download the Transmission Sphere Selection Guide
Learn how to choose the correct transmission sphere for your application.
Measuring Convex Surface Error with a Short Coherence Fizeau
Transparent domes can be challenging to measure because both surfaces of the dome reflect, creating a complex interference pattern. A short coherence AccuFiz D interferometer allows you to isolate each surface to measure independently.
Measuring Transparent Domes
Learn about measuring thin, transparent domes in this application note.
Need Assistance with Measuring Convex Surfaces?
Contact our applications staff today for expert opinions on optical measurement and on measurement equipment.
Get In Touch
3280 E Hemisphere Loop, Ste 146
Tucson, AZ 85706
Office Hours (Arizona Time)
Mon: 8am - 5pm
Tue: 8am - 5pm
Wed: 8am - 5pm
Thur: 8am - 5pm
Fri: 8am - 5pm